34972A LABVIEW DRIVERMarch 9, 2020
I copied that from a VI I created some months ago and didn’t really know how to put them together. The driver also includes a full set of functional building blocks that can be used to create a completely custom A application in LabVIEW. It displays the acquired data on a front panel indicator not pictured. But I still don’t know how to put my ‘measurement part’ and my ‘logging part’ together. The most straightforward way to take advantage of the LabVIEW instrument driver is to run the example programs, which require only basic user configuration. Thank you for the helpful response. It would be great if anyone can help me fix my problem or give me some advice how to change my VI.
|Date Added:||2 October 2009|
|File Size:||41.5 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Can I use the National Instruments 34970A LabView Driver and Examples with the 34972A?
If you are looking for an easy way to 34927a the speed, sensor support, and channel count of your test and measurement system then you may want consider adding NI CompactDAQ hardware to your setup. Message 7 of Message 10 of Introduction to the A Instrument Driver An instrument driver is a set of software routines that control a programmable instrument.
Even my data logging part works well. I changed my VI and now I can configurate the measurements.
Using the Agilent 34970A with LabVIEW
The programme works fine and I receive a text data after logging. The NI CompactDAQ platform is an excellent hardware option to consider when adding to or creating a user-defined, PC-based data logging or bench top automated test system. Message 4 of Use the instrument driver VIs to define your custom application.
The logging part seemed wrong for me too. Do I really have to use a While Loop or is there a function I don’t know yet?
Setting up the hardware to achieve these high rates, resolutions, and channel counts has never been easier or quicker.
We have used three voltage measurement configurations proposed by you at http: Bob Schor 28 August I want them to be parallelized and continuous. Most Active Software Boards: It should be around the scan and read function.
The driver also includes a full set of functional building blocks that can be used to create a completely custom A application in LabVIEW.
Thank you for the helpful response. Because creating crisp graphical user interfaces in LabVIEW is easy, we could have also chosen to visualize the data in a chart or graph. Six such ready-to-run examples labvirw provided by National Instruments in the driver. Since peak detection is one of the analysis functions available in LabVIEW, we can easily add that functionality to our application.
Continuous data acquisition with Agilent A – Discussion Forums – National Instruments
Message 2 of Can you help me fix my problem? Though all of these examples are intended to serve as a starting point for the creation of a custom application, they may be used 34972q modification to immediately control the Agilent A.
Message 3 of Here is an example of steps configuring DC volts, Scaling DC shunts, K type thermocouples, and building a scan list. It displays the acquired data on a front panel indicator not pictured. The configuration should be outside the loop.